I’ve noticed that one issue that seems to give folk concern about investing in a good Newtonian pertains to having to clean the optics every now and again. I’ve never really understood this mindset though. Having had my closed-tube 8-inch Newtonian for about 18 months now, and having clocked up a few hundred hours of observations with it, I felt it was time to give the mirrors a cleaning. Here’s how I do it:
The mirrors are removed from the tube.
First I make sure that all the loose dust and debris has been blown off using an air brush. Next, I run some cold tap water into a sink and add a drop or two of washing up liquid. The water we use here is very soft; indeed we are graced with some of the softest water in the British Isles, which also makes drinking tea especially pleasant! If your local water source is hard, I’d definitely recommend using de-ionised/distilled water.
Starting with the secondary mirror, I dip my fingers into the water and apply some of it onto the mirror surface with my finger tips, gently cleaning it using vertical strokes. Did you know that your finger tips are softer than any man-made cloth and are thus ideal for cleaning delicate surfaces like telescope mirrors?
Next, the mirror reflective surface is rinsed under some cold, running tap water.
The procedure is repeated for the primary mirror;
The mirrors are then supported on their sides to allow them to drain excess water, and then left to dry in a warm, kitchen environment. Stubborn water droplets nucleating on the mirrors are removed using some absorbent tissue.
Finally, the mirrors are placed back in the telescope tube, making sure not to over-tighten the screws which hold the primary in place inside its cell. All that remains then is to accurately align the optical train, as described previously.
There we are! Not so difficult after all; and all done in about 40 minutes! The soft water doesn’t show up any significant spots after cleaning unlike hard water sources and now the optics are as clean as the day they were produced.
With a busy season of optical testing and planetary observing ahead, I know that my 8-inch will be operating as well as it possibly can. And that’s surely good to know!
I feel a nice, hot cuppa is in order!